I understand that £15 isn't much when you look at it like that but we are just working through the desk and not even sure if it is the relay that is the problem, but we seem to swap them out quite a bit when looking for faults and as it has 36 channels that can add up to quite a lot.
We are just trying to get as much fixed ourselves before calling our tech to finish anything we can't do. Think my boss is going to give him a call to see if he has any as he has been working this desk for decades.
Originally Posted by Matt Syson
Although the 2520 and 2500 had officially been 'discontinued' when I started at AMEK I have had my head in a few. The plasma meters were one excitement!
This is the 2500 with big old VU's they tend to be too accurate, is this something that might be fixable? Or do they just need regular calibration?
Thanks for the help.